Views: 0 Author: Site Editor Publish Time: 2021-09-03 Origin: Site
During use, due to the improper size of the breathing mask, unbalanced angle adjustment, reduced elasticity, loose headgear or unbalanced pulling of the headgear on both sides, etc., air leakage will be caused, and specific problems need to be treated symptomatically.
Eyes and nose are leaking, generally consider that the forehead pad and nasal bridge arm are too high, or the lower headband is too tight. At this time, the forehead pad or the entire mask can be moved down and the headband support point can be adjusted to a lower position, or slightly loose Lower the headband; if you find air leakage under the mouth, generally consider that the forehead cushion and nasal bridge arm are too low, or the upper headband is too tight. At this time, you can move the forehead cushion or the entire mask to a higher position. Or slightly loosen the upper headband. This kind of air leakage problem can be adjusted and solved by yourself, if necessary, you need to contact a doctor for guidance.
You should contact your doctor and ventilator manufacturer in time if you encounter air leakage from your mouth or the corners of your mask.
The face mask is a consumable part of the ventilator. The silicone part will naturally age. After the ageing of the silicone, it will become hard and cause the mask to leak. It is generally recommended to change the breathing mask every 3-6 months.